I am thinking the is taking too long to make the measurement. I have not been able to find anything that provides much detail. I made up the number 50 – make sure none of the responses are longer than whatever you set! Message 1 of Hi everyone, I am a beginner of learning labview. I also tried to delete everything from keithley labview loop except the read function and an indicator of the measurement, turned auto rage off and set it to 0.
|Date Added:||7 March 2008|
|File Size:||34.41 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
With Resistance I am constantly getting an error that parameter 2 is out of range error in measurement function. Could you please help me to correct the program if something is wrong?
Keithley Picoammeter – Instrument Driver – National Instruments
Message 9 of Who is online Users browsing this forum: Unlocking them as individual files does not change this situation. This attachment does not use the internal buffer. Select Install Now you can access the VIs under the palette menu: Thank you for your reply. Hi everyone, I am a beginner of learning labview.
Or, post back if the one you think will do does. Message 9 of Otherwise why labvieq just 648 the 2x keithley labview have them source and measure at the same time?
KEITHLEY 6485 LABVIEW DRIVER
Message 2 of At the factory, the GPIB is selected. I also tried to delete everything from keithley labview loop kr the read function and an indicator of the measurement, turned auto rage off and set it to 0.
It just uses keithlej loop with a READ? Also the Picoammeter has two functions Voltage.
Solved: Labivew program for KE picoammeter – Discussion Forums – National Instruments
Message 7 of You can use only one interface at a time. Message 10 of Taking a quick look at the manual for the KeithleyI read these sections: The Keithley drivers were kekthley in the 32 bit Labview folder and I was using the 64 bit Labview program.
Besides you are lucky, because proper VISA drivers were just released for the picoamp meter very recently April! Could you please help me to correct the program if something is wrong?
Perhaps you could try this VI? I will use the picoammeter to vary the voltage while simulataneously measuring the current. N9L this blog is easy blog.
Sometimes I have difficulties when a USB to serial converter is ‘null-modem’ style – this means that it switches two of the cables inside the connector. Did you set the mode to RS?
Message 7 of I back-saved to just in case. Message 5 of For example, a simple picoamp Read test:.
It does not work. Hi Please describe which other instrument you are using.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Auto-suggest helps you quickly narrow down your 6458 results by suggesting possible matches as you type. Hi everyone, I am a beginner of learning labview.